474 Transformer Pre-amplifier
The characterisation of spectral response1 and EQE/IQE of photovoltaic devices involves the determination of device response to a monochromatic light probe in the presence of a solar simulator bias to simulate operation conditions.
In order to discriminate between the photocurrents generated by both optical stimulations, the monochromatic probe is optically chopped (AC), whilst the solar simulator is operated in the DC regime.
The 474 transformer blocks the DC component and amplifies and passes the AC component to the system lock-in amplifier.
The 474 transformer is positioned close to the device under test and coupled via low-resistance cable to ensure short circuit conditions, whilst the 474 amplifier is a single-width module housed within the 417/T mother unit.
- Core Features
- Near-short circuit device operation
- Decouples DC bias source generated photocurrent
- Signal gain provided by transformer and low-noise amplifier
1IEC60904-8, Photovoltaic devices – Part 8: Measurement of spectral response of a photovoltaic (PV) device